Want to learn more? Call 855.655.2020 or email
User Support? Call 877.393.8676 904.280.1900
Q. Has anyone had success with molding a low myope (-0.50) into a higher myope using hyperopic molding techniques in order to establish better near VA and compensate for presbyopia in a 60 y.o.
A. Attached is a design for my eye, un-molded –1.00-0.50 x095, I am 58, so need a little more help at near. My target near power is –1.75 but I wanted to minimize the distance blur. I input my rx power as +0.75, (taking me from a –1.00 to a –1.75 post mold centrally but less minus toward the peripheral). I used a thick lens, .30 center and a little loose superior and inferior to decenter the lens molding inferior thus putting the near add a little lower while still trying to get the best distance vision centrally. The lens is very comfortable. I am pleased with the vision. 20/25 distance, 20/20 near for this non dominate OS. My OD dominate I use a traditional ortho-K correction for distance. (Compliments of Dr. Terry Lawson OD, Tulsa Ok.)
Back surface of lens design
228 Canal Blvd Suite 3
Ponte Vedra FL 32082 USA
Before and after topography
Copyright © 2007-2015 Wave LLC. All Rights Reserved.